ADT's Wall Mount Breathalyser Calibration Kit comes with a Hand Simulator, Alcohol Solution, Temperature Chart with instructions and a PowerPoint Presentation on how to calibrate the ADT's Wall Mount Breathalysers.
Fill the Hand Simulator with 150ml of the provided alcohol solution. Check the temperature chart and thermometer in the Simulator to see what the expected reading is. Place the Simulator mouth piece into the sample port, pump the aspirator bulb 10 times and press the manual inject button. When the reading has stopped counting adjust the Calibration pot to the desired reading. Repeat test two more times to ensure the correct calibration has been obtained.
Write a review
Rating Bad Good